Design and examination of the new biosoluble casting alloy of the system Mg–Zr–Nd for osteosynthesis
Keywords:alloying elements, experiment design, tensile strength, relative elongation, chemical composition, optimization
We have performed a comparative analysis of existing materials for the fabrication of implants and report their physical-mechanical properties; their advantages and disadvantages have been defined. It is shown that magnesium alloys are among the most promising biosoluble materials. They are bioinert and biocompatible, but their use in osteosynthesis is limited mainly by their inadequate mechanical properties due to the high rate of biodegradation, which requires improving them by changing the chemical composition of the alloy.
In order to develop a new magnesium-based biosoluble alloy, we have selected the most suitable doping systems in accordance with the established criteria.
Employing the methods of experiment design, we studied the separate and joint influence of zirconium, neodymium and zinc on structure formation and mechanical properties of magnesium alloy. Mathematical models have been constructed that describe the influence of the examined alloying elements on the mechanical properties of the metal. Using the regression equations derived, we have carried out the optimization of the chemical composition of magnesium alloy.
The industrial and pre-clinical tests of implants made from the designed biosoluble alloy have been performed. Experiments on animals confirmed the absence of toxic effect from the products of degradation of the devised magnesium alloy on a living organism. Studying the influence of the designed alloy on reparative osteogenesis during experiment on rabbits has shown the positive dynamics of bone tissue regeneration without noticeable changes in its structure, which ensures reliable merging of elements in bones at osteosynthesis.It was established that the implants made from the designed alloy possess the necessary level of mechanical properties that match the mechanical properties of bone tissue. At the same time, they are non-toxic and provides a secure bone tissue healing until the complete fracture consolidation. Positive results of the experiments conducted allow us to suggest a favorable prognosis on the possibility of using implants made from the devised biosoluble alloy of the system Mg–Zr–Nd in humans
- Hayes, J., Richards, R. (2010). The use of titanium and stainless steel in fracture fixation. Expert Review of Medical Devices, 7 (6), 843–853. doi: https://doi.org/10.1586/erd.10.53
- Specification for Wrought Nitrogen Strengthened 21Chromium-10Nickel-3Manganese-2.5Molybdenum Stainless Steel Alloy Bar for Surgical Implants (UNS S31675) (2013). ASTM International. doi: https://doi.org/10.1520/f1586
- Wu, S., Liu, X., Yeung, K. W. K., Liu, C., Yang, X. (2014). Biomimetic porous scaffolds for bone tissue engineering. Materials Science and Engineering: R: Reports, 80, 1–36. doi: https://doi.org/10.1016/j.mser.2014.04.001
- Muzychenko, P. F. (2012). Biomaterials technology problems in traumatology and orthopedics. Travma, 1, 94–98.
- Seal, C. K., Vince, K., Hodgson, M. A. (2009). Biodegradable surgical implants based on magnesium alloys – A review of current research. IOP Conference Series: Materials Science and Engineering, 4, 012011. doi: https://doi.org/10.1088/1757-899x/4/1/012011
- Chen, C.-H., Shyu, V. B.-H., Chen, Y.-C., Liao, H.-T., Liao, C.-J., Chen, C.-T. (2013). Reinforced bioresorbable implants for craniomaxillofacial osteosynthesis in pigs. British Journal of Oral and Maxillofacial Surgery, 51 (8), 948–952. doi: https://doi.org/10.1016/j.bjoms.2013.07.011
- Vasenius, J., Vainionpää, S., Vihtonen, K., Mäkelä, A., Rokkanen, P., Mero, M., Törmälä, P. (1990). Comparison of in vitro hydrolysis, subcutaneous and intramedullary implantation to evaluate the strength retention of absorbable osteosynthesis implants. Biomaterials, 11 (7), 501–504. doi: https://doi.org/10.1016/0142-9612(90)90065-x
- Pihlajamaki, H., Bostman, O., Hirvensalo, E., Tormala, P., Rokkanen, P. (1992). Absorbable pins of self-reinforced poly-L-lactic acid for fixation of fractures and osteotomies. The Journal of Bone and Joint Surgery. British Volume, 74-B (6), 853–857. doi: https://doi.org/10.1302/0301-620x.74b6.1447246
- Lauer, G., Pradel, W., Leonhardt, H., Loukota, R., Eckelt, U. (2010). Resorbable triangular plate for osteosynthesis of fractures of the condylar neck. British Journal of Oral and Maxillofacial Surgery, 48 (7), 532–535. doi: https://doi.org/10.1016/j.bjoms.2009.10.008
- Bergsma, J. E., Bruijn, W. C., Rozema, F. R., Bos, R. R. M., Boering, G. (1995). Late degradation tissue response to poly(L-lactide) bone plates and screws. Biomoterials, 16 (1), 25–31. doi: https://doi.org/10.1016/0142-9612(95)91092-d
- Böstman, O. M. (1998). Osteoarthritis of the ankle after foreign-body reaction to absorbable pins and screws. The Journal of Bone and Joint Surgery. British Volume, 80-B (2), 333–338. doi: https://doi.org/10.1302/0301-620x.80b2.0800333
- Schumann, P., Lindhorst, D., Wagner, M. E. H., Schramm, A., Gellrich, N.-C., Rücker, M. (2013). Perspectives on Resorbable Osteosynthesis Materials in Craniomaxillofacial Surgery. Pathobiology, 80 (4), 211–217. doi: https://doi.org/10.1159/000348328
- Türesin, F., Gürsel, I., Hasirci, V. (2001). Biodegradable polyhydroxyalkanoate implants for osteomyelitis therapy: in vitro antibiotic release. Journal of Biomaterials Science, Polymer Edition, 12 (2), 195–207. doi: https://doi.org/10.1163/156856201750180924
- Chen, G.-Q., Wu, Q. (2005). The application of polyhydroxyalkanoates as tissue engineering materials. Biomaterials, 26 (33), 6565–6578. doi: https://doi.org/10.1016/j.biomaterials.2005.04.036
- Philip, S., Keshavarz, T., Roy, I. (2007). Polyhydroxyalkanoates: biodegradable polymers with a range of applications. Journal of Chemical Technology & Biotechnology, 82 (3), 233–247. doi: https://doi.org/10.1002/jctb.1667
- Berins, L. M. (2000). SPI Plastics Engineering Handbook of the Society of the Plastics Industry. Springer.
- Barinov, S. M. (2010). Calcium phosphate-based ceramic and composite materials for medicine. Russian Chemical Reviews, 79 (1), 13–29. doi: https://doi.org/10.1070/rc2010v079n01abeh004098
- Barakat, N. A. M., Khil, M. S., Omran, A. M., Sheikh, F. A., Kim, H. Y. (2009). Extraction of pure natural hydroxyapatite from the bovine bones bio waste by three different methods. Journal of Materials Processing Technology, 209 (7), 3408–3415. doi: https://doi.org/10.1016/j.jmatprotec.2008.07.040
- Heise, U., Osborn, J. F., Duwe, F. (1990). Hydroxyapatite ceramic as a bone substitute. International Orthopaedics, 14 (3). doi: https://doi.org/10.1007/bf00178768
- Qiu, H., Yang, J., Kodali, P., Koh, J., Ameer, G. A. (2006). A citric acid-based hydroxyapatite composite for orthopedic implants. Biomaterials, 27 (34), 5845–5854. doi: https://doi.org/10.1016/j.biomaterials.2006.07.042
- Li, J., Lu, X. L., Zheng, Y. F. (2008). Effect of surface modified hydroxyapatite on the tensile property improvement of HA/PLA composite. Applied Surface Science, 255 (2), 494–497. doi: https://doi.org/10.1016/j.apsusc.2008.06.067
- Shikinami, Y., Okuno, M. (1999). Bioresorbable devices made of forged composites of hydroxyapatite (HA) particles and poly-L-lactide (PLLA): Part I. Basic characteristics. Biomaterials, 20 (9), 859–877. doi: https://doi.org/10.1016/s0142-9612(98)00241-5
- Staiger, M. P., Pietak, A. M., Huadmai, J., Dias, G. (2006). Magnesium and its alloys as orthopedic biomaterials: A review. Biomaterials, 27 (9), 1728–1734. doi: https://doi.org/10.1016/j.biomaterials.2005.10.003
- Eydenzon, M. A. (1969). Magniy. Moscow: Metallurgiya, 351.
- Karpov, V. G., Shahov, V. P. (2001). Sistemy vneshney fiksacii i regulyatornye mekhanizmy optimal'noy biomekhaniki. Moscow: SST.
- Song, G. (2007). Control of biodegradation of biocompatable magnesium alloys. Corrosion Science, 49 (4), 1696–1701. doi: https://doi.org/10.1016/j.corsci.2007.01.001
- Witte, F., Kaese, V., Haferkamp, H., Switzer, E., Meyer-Lindenberg, A., Wirth, C. J., Windhagen, H. (2005). In vivo corrosion of four magnesium alloys and the associated bone response. Biomaterials, 26 (17), 3557–3563. doi: https://doi.org/10.1016/j.biomaterials.2004.09.049
- Song, G., Song, S. (2007). A Possible Biodegradable Magnesium Implant Material. Advanced Engineering Materials, 9 (4), 298–302. doi: https://doi.org/10.1002/adem.200600252
- Samarskiy, A. A., Mihaylov, A. P. (2001). Matematicheskoe modelirovanie. Moscow: Fizmatlit, 320.
- Velikiy, V. I., Yares’ko, K. I., Shalomeev, V. A., Tsivirko, E. I., Vnukov, Y. N. (2014). Prospective Magnesium Alloys with Elevated Level of Properties for the Aircraft Engine Industry. Metal Science and Heat Treatment, 55 (9-10), 492–498. doi: https://doi.org/10.1007/s11041-014-9660-x
- Shalomeev, V., Tsivirco, E., Vnukov, Y., Osadchaya, Y., Makovskyi, S. (2016). Development of new casting magnesium-based alloys with increased mechanical properties. Eastern-European Journal of Enterprise Technologies, 4 (1 (82)), 4–10. doi: https://doi.org/10.15587/1729-4061.2016.73384
- GOST 2856-79. Splavy magnievye liteynye. Marki (1981). Moscow: Izdatel'stvo standartov.
- Xingwei, Z., Jie, D., Wencai, L., Wenjiang, D. (2011). Microstructure and mechanical properties of NZ30K alloy by semicontinuous direct chill and sand mould casting processes. China foundry, 8 (1), 41–46.
- Turowska, A., Adamiec, J. (2015). Mechanical Properties of WE43 Magnesium Alloy Joint at Elevated Temperature / Właściwości Mechaniczne Złączy Ze Stopu Magnezu WE43 W Podwyższonej Temperaturze. Archives of Metallurgy and Materials, 60 (4), 2695–2702. doi: https://doi.org/10.1515/amm-2015-0434
- Gill, L., Lorimer, G. W., Lyon, P. (2005). Microstructure/Property Relationships of Three Mg-RE-Zn-Zr Alloys. Magnesium, 421–426. doi: https://doi.org/10.1002/3527603565.ch66
- Koltygin, A. V. (2013). Analiz vozmozhnyh fazovyh prevrashcheniy pri kristalizacii i ih vliyanie na liteynuyu strukturu v splave ML10. Metallovedenie i termicheskaya obrabotka metallov, 8, 25–28.
How to Cite
Copyright (c) 2019 Vadim Shalomeev, Nikita Aikin, Vadim Chorniy, Valeriy Naumik
This work is licensed under a Creative Commons Attribution 4.0 International License.
The consolidation and conditions for the transfer of copyright (identification of authorship) is carried out in the License Agreement. In particular, the authors reserve the right to the authorship of their manuscript and transfer the first publication of this work to the journal under the terms of the Creative Commons CC BY license. At the same time, they have the right to conclude on their own additional agreements concerning the non-exclusive distribution of the work in the form in which it was published by this journal, but provided that the link to the first publication of the article in this journal is preserved.
A license agreement is a document in which the author warrants that he/she owns all copyright for the work (manuscript, article, etc.).
The authors, signing the License Agreement with PC TECHNOLOGY CENTER, have all rights to the further use of their work, provided that they link to our edition in which the work was published.
According to the terms of the License Agreement, the Publisher PC TECHNOLOGY CENTER does not take away your copyrights and receives permission from the authors to use and dissemination of the publication through the world's scientific resources (own electronic resources, scientometric databases, repositories, libraries, etc.).
In the absence of a signed License Agreement or in the absence of this agreement of identifiers allowing to identify the identity of the author, the editors have no right to work with the manuscript.
It is important to remember that there is another type of agreement between authors and publishers – when copyright is transferred from the authors to the publisher. In this case, the authors lose ownership of their work and may not use it in any way.